How to solve intraoral camera troubles?
If there are some troubles of intraoral camera, Please see below special case firstly. If still can’t be solved, please contact us.
|1||the products can't be startup when open (ON/OFF)switch)||1.check the power socket if it is ok
2.check the cable if it is connected
|2||computer or TV can't reveal images||1.check the connection between products and computer or TV if it is correct
2.check the Driver Setup if it is finished,
|3||in the process of use, the image will be a little blurry||If you use it in the first time, it belongs to natural reaction. It will be better later on|
|4||freezing images then appears dithering||the hand shakes when uses camera to capture image, when users are skillful for use products, the phenomena can be changed|
|5||the images of computer or TV have spots||1.check the screen if it has spot
2.check the lens if it has spot
3.if the spot can't be cleaned, please contact the local dealer
Notice: Before each usage, check the outer surface of the intraoral camera for any signs of physical damage or defect. The surface of the intraoral camera should have a smooth finish, with no evidence of chipping or damage.
KNOWLEDGE OF WARNINGS AND CAUTIONS
Users must exercise every precaution to ensure personnel safety, and be familiar with the warnings. And cautions presented throughout this manual and summarized below.
CAUTION: TO REDUCE THE RISK OF ELECTRICAL SHOCK, DO NOT REMOVE THE COVER OF THE CONTROL MODULE OR POWER SUPPLY ADAPTER. THERE ARE NO USER-SERVICEABLE PARTS INSIDE. SERVICING SHOULD BE PERFORMED BY QUALIFIED DEALER SERVICE REPRESENTATIVES ONLY.CONNECTING ANY DEVICE TO THE SUPER CAMERA SYSTEM THAT DOES NOT MEET THE EQUIVALENT SAFETY REQUIREMENTS OF THE SYSTEM MAY REDUCE THE SAFETY EFFECTIVENESS OF THE SUPER INTRAORAL CAMERA SYSTEM.
TURN OFF POWER AND UNPLUG THE SUPER INTRAORAL CAMERA SYSTEM COMPONENTS BEFORE PERFORMING CLEANING PROCEDURES.
WARNING: TO PREVENT FIRE OR ELECTRICAL SHOCK, DO NOT EXPOSE THIS EQUIPMENT TO RAIN OR MOISTURE.
DO NOT SPRAY CLEANING LIQUIDS OR DISINFECTANTS DIRECTLY ON THE INTRAORAL CAMERA HANDPIECE OR CONTROL MODULE. USE CARE NOT TO ALLOW LIQUIDS TO RUN INTO INTERNAL CIRCUITRY.
DO NOT WIPE THE SURFACES USING BENZINE, THINNER, ETC. AS THIS MAY DEGRADE THE FINISH.